Plug 9V power,integrated cable tester is auto start-up and starts to buzzing. It shows“tst: 00” on the screen. “00” position will in turn fast show serial number of position of testing wire
access to instrument. “-3 -” indicates that select stored third detection data of the tester,when testing wires.
2. Setting the line sequence of the test wires: Connect testing wire to the tester and long press D, enter into the learning mode. For example, “-3 -” blink in digital that detection data will be stored in the tester.Long press C to switch the instrument storage test number.The instrument can reserve up to nine kinds of detection data. ClickAor B to start learning, the buzzer send beeps, then
the learning end.
3. Testing on wires: After connect wires to the tester,clickA or B to start to test the wire.
When a testing wire with problems, the screen will show “Err: XX” and the buzzer sounds consecutive warning drops. If it shows “Err: 15”, which indicates No. 15 wire abnormal. PressA or B to exit. When the testing wire is qualified, the
buzzer short call, and “PASS” is instrumented on the detector.
4. The test ends: Unplug the power of the detector; make a note of line sequence of wires connected to the detector and the instrument storage testing number for testing this kind of wires later.Collated test line, the test is complete.
5. Detect storage line sequence: When power on, and long press C to select instrument storage testing number.The selected number is
blinking, click theAor B to confirm. The wires are contacted to the tester correspondingly as previously storage line sequence records. Then follow the third step to test wires.